Quick visual check for additional safety: (= stability check)The round control window on the back of the test strip must not differ from the reference colour “0” on the colour scale on the test strip container. If the round control window has turned a different colour, the test strip is unusable and should be discarded.
Quick visual reading back-up for additional safety: It is possible to verify the result by comparing the colour of the round control window on the back of the test strip with the colour scale on the label of the test strip container-especially in case of implausible meter readings.
Visual underdosing: Detection for safe test results Since the test pad changes colour once the blood has been applied, the user can recognize if too little sample has been applied. The centre of the test pad must be uniformly covered. The control window on the back of the test strip must be completely discoloured.
Optimal test strip architecture for hygienic handling: The size and the architecture of the test strip allow test strip handling without coming into contact with the blood specimen-especially when taking the used test strip off the monitor.
Large test pad for easy blood application: The large test pad can be touched and accurate application is easy.
Rapid blood absorption: Quick blood absorption makes the test hygienic and fast. The test strip needs only 1-2 µL but the test pad can absorb up to 30 µL thanks to the reservoir for excessive blood!
Sample spreads automatically: Thanks to the test strip mesh, the sample is spread evenly and very quickly over the test pad. This makes1-2 µL of sample sufficient for performing the test.
Handy top dosing: The large test pad is easy to find and to fill even for people who are visually or physically impaired. Also, special situations (e.g. hypos) do not cause handling problems.
Easy to handle: The large size of the test strip itself makes it easy to grasp, hold, insert into the meter and discard.
Flexible – thanks to out-of-meter dosing and AST options: The Accu-Chek Active test strip can be dosed outside the monitor. This and the fact that only1-2µL blood are required, make AST testing comfortable and convenient.
HOW TO USE
- Clean the finger before use.
- Piercing the finger from where you want to take the blood sample.
- The minimum blood sample should be 0.3 mg/dl.
- Put the sample blood on the test strip and use it as directed in the